POSITIONING THE STEM Before installation,the stem should be set to the desired angle as follows:
Figure 1: Loosen the four screws labeled“A” and “B” in Figure 1, until the harness revolves freely without twisting theflexible housing.
Figure 2: While holding the case, revolvethe harness clockwise or counterclockwise, as indicated by arrows in Figure 2,to place the harness in a position that will permit flexing the stem in thedesired direction with respect to the case. Then lock the two screws labeled“A.”
Figure 3: Flex the stem to the desiredangle with respect to the face of the thermometer, as shown in Figure 3, thenlock the two screws labeled “B.”
To assure longest life, the “EVERY- ANGLE”joint should be operated only when necessary during installation or removal ofthe thermometer.